Welcome to the Podiatry Arena forums, for communication between foot health professionals about podiatry and related topics.
You are currently viewing our podiatry forum as a guest which gives you limited access to view all podiatry discussions and access our other features. By joining our free global community of Podiatrists and other interested foot health care professionals you will have access to post podiatry topics (answer and ask questions), communicate privately with other members (PM), upload content, view attachments, receive a weekly email update of new discussions, earn CPD points and access many other special features. Registered users do not get displayed the advertisments in posted messages. Registration is fast, simple and absolutely free so please, join our global Podiatry community today!
If you have any problems with the registration process or your account login, please contact contact us.
Preschool-age children with neuromuscular disorders are often excluded from clinical trials due to the lack of reliable and objective strength measures. We evaluated the reliability of measuring foot and ankle muscle strength in 60 healthy children age 2-4 years. The strength of foot inversion and eversion, ankle plantarflexion, and dorsiflexion was measured using a hand-held dynamometer. Intrarater and interrater reliability of two assessors was determined for each muscle group using intraclass correlation coefficients (ICCs), 95% confidence intervals (CIs), and standard error of measurement (SEM). For all muscle groups intrarater (ICC(2,2) = 0.85-0.94, 95% CI = 0.75-0.96, SEM = 1.5-4.7 N) and interrater (ICC(2,1) = 0.88-0.96, 95% CI = 0.81-0.98, SEM = 1.2-4.6 N) reliability was high for all children. As expected, reliability was generally highest in 3- and 4-year-old children and lowest in 2-year-old children. Hand-held dynamometry can reliably measure foot and ankle strength in very young children and may help aid in diagnosis